Some customers suspect the accuracy of the WT2080 because there is some differences of the test result between LISUN WT2080 LED Power Driver Tester and multimeter when testing LED power driver. In fact, the test results are different because the test principle is different: the WT2080 test result is the real effective value (RMS), the multimeter test result is the average value.
LISUN WT2080 is according to GB/T 24825-2009 and IEC 62384:2006. It can measure the following parameters: Input Characteristics Test (AC/DC), Output Characteristics Test (AC/DC), Start Characteristics Test (DC) and Hormanic Test. It is specified in the standard that the test result should be RMS.
The results of the general multimeter test are average values, so the test results are naturally different because of the different standards. WT2080 can print data and communicate with PC, displaying all the dates and curves in PC. There is also the average value in the test report, but on the display screen is RMS. Click here to learn more about the WT2080 LED Power Driver Tester.
Lisun Instruments Limited was found by LISUN GROUP in 2003. LISUN quality system has been strictly certified by ISO9001:2015. As a CIE Membership, LISUN products are designed based on CIE, IEC and other international or national standards. All products passed CE certificate and authenticated by the third party lab.
Our main products are Goniophotometer, Spectroradiometer, Integrating Sphere, LED Test Instruments, CFL Test Instruments, Photometer and Colorimeter, EMC & EMI Testing, Surge Generator, Electrical Safety Testing, Environmental Test Chamber, Waterproof Test , Dustproof Test Chamber, High and Low temperature Humidity Chamber, Plug and Switch Testing, AC and DC Power Supply.